1 Sterilisation Wrap COMPLIANCE TO. EN ISO 11607 -1:2006/. AMD 1:2014 . 1. The manufacturer, Halyard Health, does not accept any responsibility for the incorrect choice or misuse of the product shown in this brochure. All information contained in this brochure is as accurate as possible at the time of publication, however legislation and regulations are under constant review and may change in the lifetime of this brochure. Accordingly the specification for the product may be subject to change.. 2 3. TABLE OF CONTENTS HALYARD* STERILISATION WRAP. (NUMBERING FOLLOWS THE ORDER OF EN ISO 11607 -1:2006) LIST OF USED ACRONYMS AND ABBREVIATIONS.
2 INTRODUCTION. 4. GENERAL REQUIREMENTS. Quality Systems Sampling Test Methods Documentation 5. MATERIALS AND PREFORMED STERILE BARRIER SYSTEMS C : Degree Celsius l: Liter General Requirements AAMI : Association for the Advancement of Medical Instrumentation lbs : Pounds AATCC : American Association of Textile Chemists and Colorists LPS : Laser Particle Counter Microbial Barrier Properties ANSI : American National Standards Institute m2 : Square meter Compatibility With The Sterilisation Process ASP : Advanced Sterilisation Products mbar : Millibar Compatibility With The Labelling System ASTM : American Society for Testing and Materials MD tear : Machine Direction Storage and Transport AVG BW.
3 Average Basis Weight mg : Milligram BFE : Bacterial Filtration Efficiency ml : Milliliters Appendix 1: ISO 13485: 2003 Certificate BI : Biological Indicator mm : Millimeter Appendix 2: Bioburden Certificate CD tear : Cross Direction M : Micrometer Appendix 3: Physical Properties cfm : Cubic feet per minute MPI : Maintenance of package integrity CFU : Colony Forming Units NFPA : National Fire Prevention Association Appendix 4: Chemical Properties cm2 : Square centimeter osy : Ounces per square yard Appendix 5: Ethylene Oxide Sterilant Penetration And Residuals CV : Coefficient of Variation pH : Measure of the acidity or alkalinity Study Results For HALYARD* Sequential and ONE-STEP* DIN : Deutsches Institut f r Normung ppm : Part per million Sterilisation Wrap ECH : Ethylene Chlorohydrin PVC : Polyvinyl chloride Appendix 6: Pre-Vacuum Steam Sterilant Penetration Study Results For EN : European Norm RH : Relative Humidity HALYARD* Sequential and ONE-STEP* Sterilisation Wrap EO : Ethylene Oxide SAL : Sterility Assurance Level F: Fahrenheit SBS : Sterile Barrier System Appendix 7: Formaldehyde Sterilisation Compatibility And Residuals FIFO : First In First Out sec.
4 Seconds Study Results For HALYARD* Sequential and ONE-STEP*. FTMS : Flexible Test and Measurement System SOPs : Standard Operating Procedures Sterilisation Wrap in2 : Square inches STD : Standard Deviation Appendix 8: Sterilucent Sterilisation Compatibility And MPI Study Results INDA : International Nonwovens and Disposables Association TC : Technical Committee For HALYARD ONE-STEP* Sterilisation Wrap ISO : International Organization for Standardization TNO : Netherlands Organization for Applied Scientific Research Appendix 9: Sterrad Sterilisation MPI Study Results for HALYARD IST : International Standard Test UV.
5 Ultraviolet ONE-STEP* and QUICK-CHECK* Sterilisation Wrap kg : Kilogram Appendix 10: Amsco V-Pro Sterilisation MPI Study Results Of HALYARD. ONE-STEP* Sterilisation Wrap Appendix 11: E. O and Pre-Vacuum Steam Sterilisation MPI Study Results for HALYARD* Sequential and ONE-STEP* Sterilisation Wrap Appendix 12: Basis Weight Of Materials Not Requiring Conditioning Appendix 13: Executive Summary Colorfastness Study For HALYARD. ONE-STEP* Sterilisation Wrap Appendix 14: Final Pack Test Method for HALYARD ONE-STEP*. Sterilisation Wrap Using the Prion Cycle (18 minutes). 4 5. HALYARD* STERILISATION WRAP. COMPLIANCE TO EN ISO 11607 -1:2006.
6 4. GENERAL REQUIREMENTS. Quality systems T. he activities described in this part of EN ISO 11607 shall be carried out within a formal quality system. HALYARD* Sterilisation products are manufactured in our US facility. This facility is certified by the following documents: See Appendix 1: ISO 13485: 2003 certificate INTRODUCTION Sampling Dear Customer, The sampling plans used for selection and testing of packaging systems shall be applicable to packaging In July 2014, the technical committee ISO/TC 198 (Sterilisation of health care products) systems being evaluated. Sampling plans shall be based upon statistically valid rationale.
7 Published the amendment of EN ISO 11607 -1. All testing relative to COMPLIANCE to EN ISO 11607 -1 was conducted on product randomly selected from The major amendments to EN ISO 11607 -1 are the altered definition of a microbial barrier. distribution and thus representative of normal variations. The 2014 amendment of EN ISO 11607 -1 refers to a microbial barrier as the property of Test methods the sterile barrier system which ensures that it prevents the ingress of microorganisms, A. ll test methods used to show COMPLIANCE with this International Standard shall be validated and demonstrated under test conditions which consider sterilisation process, handling, documented.
8 Distribution, transport and storage. Test method Test Halyard Health always considered the new definition as being part of providing a proper ISO 6588-2 Colour leach microbial barrier; hence MPI (Maintenance of Package Integrity) testing has been performed ASTM D3776 6M Grammage on HALYARD* Sterilisation products prior to the revision of EN ISO 11607 -1. INDA Standard Test IST (01) Gelbo Lint This document should resolve most of your questions. However, if you do have additional ISO 1974 MD Tear questions, please contact your local Halyard sales representative. This document lists each ISO 1974 CD Tear requirement of EN ISO 11607 -1, which is followed by COMPLIANCE explanation for the relevant ISO 2758 Bursting Strength (dry).
9 Clause. The numbering is done according to the EN ISO 11607 -1's clauses. ISO 3689 Bursting Strength (wet). EN ISO 1924-2 Elongation (MD). EN ISO 1924-2 Elongation (CD). ISO 1924-2 MD Tensile Strength (dry). ISO 1924-2 CD Tensile Strength (dry). ISO 3781 MD Tensile Strength (wet). ISO 3781 CD Tensile Strength (wet). ISO 6588-2 pH. ISO 9197 Sodium Chloride content ISO 9198 Sodium Sulphate content DIN 58953-6:2010 Fluorescence 5514 FTMS No. 191A, INDA -92 Hydrostatic Head Pressure ASTM F2101 Bacterial Filtration Efficiency Final Pack Method TNO Microbial Barrier 6 7. Test method validation shall demonstrate the suitability of the methods as used.
10 The following b) pressure range elements shall be included: HALYARD* Sterilisation Wrap is not impacted by variations in pressure differences during normal establishment of a rationale for the selection of the appropriate tests for the packaging system; conditions of use. establishment of acceptance criteria; c) humidity range determination of test method repeatability; EO sterilisation should be performed at 40% - 80% relative humidity. determination of test method reproducibility; The ideal humidity range immediately prior to use is ranging from 30% to 60%. establishment of test method sensitivity for integrity tests.