1 owner 's Operator and Maintenance Manual Invacare Continuing Care, Inc. Carroll DLX Series Long Term Care Beds IH820 DLX. IH820-3 MDLX. SC900 DLX. DEALER: This Manual MUST be given to the user of the product. USER: BEFORE using this product, read this Manual and save for future reference. For more information regarding Invacare Continuing Care, Inc. (ICCI) products, parts, and services, please visit or call 800-668-2337. WARNING. DO NOT use this product or any available optional equipment without first completely reading and understanding these instructions and any additional instructional material such as owner 's manuals, service manuals or instruction sheets supplied with this product or optional equipment. If you are unable to understand the warnings, cautions or instructions, contact a healthcare professional, dealer or technical personnel before attempting to use this equipment - otherwise, injury or damage may occur.
2 The initial set up of this bed MUST be performed by a qualified technician. Procedures other than those described in this Manual MUST be performed by a qualified technician. ACCESSORIES WARNING. ICCI products are specifically designed and manufactured for use in conjunction with ICCI accessories. Accessories designed by other manufacturers have not been tested by ICCI and are not recommended for use with ICCI products. SPECIAL NOTE. For your convenience, the January 2008 version of the FDA's bed safety guidelines are provided in the Appendix. The information from the FDA's brochure, published by Hospital Bed Safety Workgroup, is reproduced verbatim, the latest revision of which is available at Carroll DLX Series 2 Part No 1134871. TABLE OF CONTENTS. TABLE OF CONTENTS. SPECIAL NOTES .. 5. TYPICAL PRODUCT PARAMETERS .. 6. LABEL LOCATIONS .. 7. SECTION 1 GENERAL GUIDELINES .. 8. Operation Radio Frequency Interference.
3 9. Weight SECTION 2 PACKAGING, HANDLING AND SET-UP .. 11. Unpacking ..11. Set-Up ..12. Inspecting the Connecting the Pendant ..13. Storage ..14. SECTION 3 OPERATION .. 15. Electric Manual Rolling the Bed ..16. SECTION 4 OPTIONAL ACCESSORIES .. 17. Bed Rails and Positioning Devices ..17. Positioning the Head/Foot Board Mounting Brackets ..17. Installing/Removing the Head/Foot Boards ..18. Installing Bolt On Head/Foot Boards ..18. Removing Bolt On Head/Foot Boards ..19. Installing Quick-Release Head/Foot Removing Quick-Release Head/Foot Boards ..19. Installing Mattress Keepers ..19. Using Emergency Installing Bed Extender - Perforated Pan Installing Vinyl Bumpers with IV Rod SECTION 5 TROUBLESHOOTING AND Maintenance .. 23. Cleaning the Beds ..23. SECTION 6 REPAIR AND REPLACEMENT .. 25. Removing/Installing the Junction Removing ..25. Part No 1134871 3 Carroll DLX Series TABLE OF CONTENTS. TABLE OF CONTENTS.
4 Replacing Head, Foot, or Hi/Lo Motor ..28. Replacing Head Motor ..28. Replacing the Foot Replacing the Hi/Lo Motor ..30. APPENDIX .. 32. A Guide to Bed Safety Bed Rails in Hospitals, Nursing Homes and Home Health Care: The Facts ..32. Bed Rail Entrapment Statistics ..32. Patient The Benefits and Risks of Bed Meeting Patients' Needs for Safety ..33. Which Ways of Reducing Risks are Best?..33. Patient or Family Concerns About Bed Rail Use ..34. Developed by the Hospital Bed Safety Workgroup ..34. LIMITED WARRANTY .. 35. Carroll DLX Series 4 Part No 1134871. SPECIAL NOTES. SPECIAL NOTES. Signal words are used in this Manual and apply to hazards or unsafe practices which could result in personal injury or property damage. Refer to the table below for definitions of the signal words. SIGNAL WORD MEANING. Danger indicates an imminently hazardous situation which, if not avoided, will DANGER. result in death or serious injury.
5 Warning indicates a potentially hazardous situation which, if not avoided, could WARNING. result in death or serious injury. Caution indicates a potentially hazardous situation which, if not avoided, may CAUTION. result in property damage or minor injury or both. NOTICE. THE INFORMATION CONTAINED IN THIS DOCUMENT IS SUBJECT TO. CHANGE WITHOUT NOTICE. PENDANT OPERATION. A safety feature of this product includes protection against overheating caused by excessive or extended periods of operation. Depending on the duration, this includes multiple or repeated adjustments or the use of multiple functions at one time. To ensure trouble free operation, ALWAYS allow a slight pause between multiple adjustments and avoid pressing more than one function button at a time. If thermal protection activation should occur, the bed will not respond to pendant commands. Given this situation, release the pendant button and allow the bed unit to sit for several minutes.
6 This will allow the protection function time to reset and restore bed function. Depending on severity of the initial overheating, this could take up to 30 minutes. Check all parts for shipping damage and test before using. In case of damage, DO. NOT use. Contact ICCI Customer Service Department (1-800-668-2337)/Carrier for further instruction. This Manual /Electric bed has been engineered to provide reliable operation. The bed has been thoroughly tested and inspected prior to shipment. This product has been supplied from an environmentally aware manufacturer that complies with the Waste Electrical and Electronic Equipment (WEEE) Directive 2002/96/CE. This product may contain substances that could be harmful to the environment if disposed of in places (landfills) that are not appropriate according to legislation. Please be environmentally responsible and recycle this product through your recycling facility at its end of life.
7 Part No 1134871 5 Carroll DLX Series Typical Product Parameters TYPICAL PRODUCT PARAMETERS. REGULATORY LISTING: UL 60601-1, IEC 60601-2-38, CAN/CSA No. 601-1. ELECTRICAL REQUIREMENTS: 120 VAC, 50-60 Hz RATED CURRENT INPUT: A max DUTY CYCLE: 10%, max 2 min/18 min RECOMMENDED STORAGE AND. SHIPPING TEMPERATURE: -29 C to 65 C (-20 F to 150 F). Degree of protection against electric shock, Type B equipment Attention - Consult Accompanying Documents IP66 International Protection Code Degree of protection against solid foreign objects and ingress of water. Type of protection against electric shock. Class II, Double Insulation IH820 DLX AND SC900 DLX. IH820-3 MDLX. OVERALL WIDTH: 35 inches (36 inches with head/foot boards). OVERALL LENGTH: 79 inches (82 inches with head/foot boards). ADJUSTMENT RANGE: 13 3/ to 27 inches 9 7/8 to 20 1/4 inches 4. UNDER BED CLEARANCE. BED LOWERED: 7 inches 41/2 inches BED ELEVATED: 8 1/4 inches 5 inches HEAD ANGLE RANGE: 0 to 70 degrees 0 to 70 degrees FOOT ANGLE RANGE: 0 to 29 degrees 0 to 29 degrees APPROXIMATE OVERALL.
8 WEIGHT (IN POUNDS): 155 - Manual Hi/Lo 170 - Electric Hi/Lo 170 - Electric Hi/Lo (3 Motors) (3 Motors). APPROXIMATE SHIPPING. WEIGHT (IN POUNDS): 165 - Manual Hi/Lo 180 - Electric Hi/Lo 180 - Electric Hi/Lo (3 Motors) (3 Motors). WEIGHT LIMITATION: 450 lbs* (205 Kgs) 450 lbs* (205 Kgs). NOTE: * Includes all accessories. Carroll DLX Series 6 Part No 1134871. LABEL LOCATIONS. LABEL LOCATIONS. NOTE: Top view of bed shown. CAUTION. KEEP SWITCH. CORD CLEAR OF. MOVING PARTS. Label P/N - 1118366 - Rev A 03/03. OF CHILDREN. OUT OF REACH. USE AND STORE. WHEN NOT IN. PENDANT. REMOVE. WARNING. CAUTION. CLEANING INSTRUCTIONS. See owner 's Manual 1. Unplug product before washing. 2. Make sure all electrical parts (motors, junction box and pendant) are not damaged and that all connection ports are plugged. WARNING. 3. Disinfect and wash all components. REMOVE. P/N 1116654. 4. DO NOT power wash or steam clean.
9 Use household water pressure. PENDANT. 5. Rinse thoroughly with water (maximum temperature 110 ). WHEN NOT IN. 6. Dry all components before storing. USE AND STORE. OUT OF REACH. OF CHILDREN. Label P/N - 1118366 - Rev A 03/03.. )"0 1, !2 -1")1# ), +"!% )# '(! !"!#$! 1, ,-&..&/-! &#!"'(!')$" .. )"*+! % !2 -1")1# #1 &" !. -13! !#. !"!#$! % &#!"'(!")*+!" )#,-&..&/-!" .. Model No. Invacare Continuing Care, Inc. Sanford, Florida 32773 Volts Hz Amps ATTENTION Conforms to UL 60601-1. ! CONSULT ACCOMPANYING. DOCUMENTS. Certified to IEC 60601-2-38. and CAN/CSA No. Serial No. Duty Cycle: Max10%, max 2min/18min IP66. 74103 P/N 1132007 Rev C. Part No 1134871 7 Carroll DLX Series SECTION 1 GENERAL GUIDELINES. SECTION 1 GENERAL GUIDELINES. WARNING. SECTION 1 - GENERAL GUIDELINES contains important information for the safe operation and use of this product. Operation Information When using an extension cord, use only a three wire extension cord having at least 16.
10 AWG (American Wire Gauge) wire and the same or higher electrical rating as the device being connected. Use of improper extension cord could result in a risk of fire and electric shock. Three prong to two prong adapters should not be used. Use of three prong adapters can result in improper grounding and present a shock hazard to the user. After installation and before use make sure all attaching hardware is securely tightened. DO NOT use near explosive gases. Possible fire hazard when used with oxygen administering equipment other than nasal or masked type. When using nasal or masked type administering equipment, oxygen or air tubing MUST. be routed and secured properly to ensure that tubing does not become entangled, kinked or severed during normal operation of Manual /Electric bed. Periodically inspect all electrical components for damage and replace if found. Continued use of a damaged component can result in abnormal operation and may present a hazard to some users.