1 Product data SheetSeptember 201700813-0100-4690, Rev PERosemount 2088 Absolute and Gage pressure transmitter Performance of with high accuracy option Lightweight, compact design for cost-effective installation Protocols available include 4 20 mA HART and 1 5 Vdc HART Low Power Absolute and gage pressure ranges up to 4,000 psi (276 bar) Rangeability of 50:12 Rosemount 2088 September 2088 pressure transmitter Product offeringProven reliability for gage and Absolute applications Available protocols include 4 20 mA HART and 1 5 Vdc HART Low Power Fully configurable LCD display to display process variable, percent of range, and diagnostic messages Lightweight, compact design enables easy installation Choice of stainless steel or Alloy C-276 wetted materialsUnlock the value of devices with the Emerson Wireless THUM Adapter Gain access to field intelligence and improve quality, safety, availability, operations, and maintenance costs Remotely manage devices and monitor health Enable new wireless measurement points Utilize existing loop powerProven, reliable, and innovative DP Level technologies Connect to virtually any process with a comprehensive offering of process connections, fill fluids.
2 Direct mount or capillary connections and materials Quantify and optimize total system performance with QZ optionInstrument manifolds quality, convenient, and easy Designed and engineered for optimal performance with Rosemount transmitters Save installation time and money with factory assembly Offers a variety of styles, materials, and configurationsContentsOrdering information .. 3 Specifications .. 7 Product Certifications .. 11 Dimensional drawings.. 16 Options .. 183 Rosemount 2088 September informationRosemount 2088 In-line pressure transmitter Specification and selection of Product materials, options, or components must be made by the purchaser of the equipment. See page 9 for more information on material output code4 20 mA HARTR osemount 2088 with Selectable HARTS1 5 Vdc Low PowerRosemount 2088 with Selectable HARTNA dditional informationSpecifications: page 7 Product Certifications: page 11 Dimensional drawings: page 16 Table 1.
3 Rosemount 2088 pressure transmitter Ordering InformationThe starred offerings ( ) represent the most common options and should be selected for best delivery. The non-starred offerings are subject to additional delivery lead description2088 pressure transmitter Measurement typeAAbsolute GGage pressure ranges Rosemount 2088 GRosemount2088A1 to 30 psi ( 1,01 to 2,1 bar)0 to 30 psi (0 to 2,1 bar) 2 to 150 psi ( 1,01 to 10,3 bar)0 to 150 psi (0 to 10,3 bar) 3 to 800 psi ( 1,01 to 55,2 bar)0 to 800 psi (0 to 55,2 bar) 4 to 4,000 psi ( 1,01 to 275,8 bar)0 to 4,000 psi (0 to 275,8 bar) Trans m itter o u tp u t(1)S4 20 mA dc/Digital HART Protocol N1 5 Vdc Low Power/Digital HART Protocol Materials of construction(2)Process connectionIsolating diaphragmFill fluid22316L SST316L SSTS ilicone 33 Alloy C-276 Alloy C-276 Silicone 2B316L SST316L SSTI nert4 Rosemount 2088 September connectionA1/2 14 NPT female B(3)DIN 16288 G 1/2 male D(3)(4)M20 male C(3)(4)RC 1/2 femaleConduit entry11/2 14 NPT 2(3)M20 4(3)
4 (5)G 1/2 Options (Include with selected model number)Extended Product warrantyWR33-year limited warranty WR55-year limited warranty Diaphragm seal assemblies(6)(7)S1 Assemble to one Rosemount 1199 Diaphragm Seal Display and interfaceM4 LCD display with local operator interface M5 LCD display, configured for engineering units Configuration buttonsD4 Analog zero and span DZdigital zero trim Mounting bracketsB4 SST mounting bracket with SST bolts Product certificationsC6 Canada Explosionproof, Intrinsic Safety, Division 2, and Dust-Ignitionproof E2 INMETRO Flameproof E3 China Flameproof E4(3)(8)Japan Flameproof E5 USA Explosionproof (XP) and Dust-Ignitionproof (DIP) E7 IECEx Flameproof EDAT E X F la m ep r o o f EMTechnical Regulations Customs Union (EAC) Flameproof I1(3)ATEX Intrinsic Safety I2 INMETRO Intrinsic Safety Table 1. Rosemount 2088 pressure transmitter Ordering InformationThe starred offerings ( ) represent the most common options and should be selected for best delivery.
5 The non-starred offerings are subject to additional delivery lead 2088 September Intrinsic Safety I5 USA Intrinsic Safety (IS) and Nonincendive (NI) I7 IECEx Intrinsic Safety IMTechnical Regulation Customs Union (EAC) Intrinsic Safety K1 ATEX Flameproof, Intrinsic Safety, Type n, Dust K2 INMETRO Flameproof, Intrinsic Safety K5 USA Explosionproof, Dust-Ignitionproof, Intrinsic Safety, and Nonincendive K6(3)Canada Explosion-Proof, Intrinsic Safety, Division 2, Dust-ignitionproof, and ATEX Flameproof, Intrinsic Safety K7 IECEx Flameproof, Intrinsic Safety, Type n, Dust KBUSA Explosionproof, Dust-Ignitionproof, Intrinsic Safety, Nonincendive, and Canada Explosionproof, Dust-Ignitionproof, Intrinsic Safety, Division 2 KMTechnical Regulation Customs Union (EAC) Flameproof and Intrinsic Safety KH(3)ATEX Flameproof, Intrinsic Safety, and USA Explosion-Proof, Dust Ignitionproof, Intrinsic Safety, and Nonincendive N1(3)ATEX Type n N3 China Type n N7 IECEx Type n ND(3)AT E X D us t NKIECEx Dust Shipboard approvalsSBSA merican Bureau of Shipping (ABS) Type Approval SBVB ureau Veritas (BV) Type Approval SDNDet Norske Veritas (DNV) Type Approval SLLL loyd's Register (LR) Type Approval pressure testingP1 Hydrostatic testingTerm i nal b l oc ksT1 Transient protection Special cleaningP2 Cleaning for special serviceCalibration certificateQ4 Calibration certificate Quality calibration certificate traceability certificationQ8 Material traceability certification per EN 10204 Q15 Certificate of compliance to NACE MR0175/ISO 15156 for wetted materials Q25 Certificate of compliance to NACE MR0103 for wetted materials Table 1.
6 Rosemount 2088 pressure transmitter Ordering InformationThe starred offerings ( ) represent the most common options and should be selected for best delivery. The non-starred offerings are subject to additional delivery lead 2088 September signal C4(3)NAMUR alarm and saturation levels, high alarm CN(3)NAMUR alarm and saturation levels, low alarm C5 (9)(10)Custom alarm and saturation levels, high alarm (requires C9 and Configuration data sheet ) C7(9)(10)Custom alarm and saturation levels, low alarm (requires C9 and Configuration data sheet ) C8(10)Low alarm (Standard Rosemount alarm and saturation levels) Conduit plug(11)DO316 SST conduit plug ConfigurationC9 Software configuration Manifold assemblies(6)(7)S5 Assemble to Rosemount 306 Integral Manifold Calibration accuracy(12) accuracy to 10:1 turndown Water approval(13)DWNSF drinking water approval Surface finishQ16 Surface finish certification for sanitary remote seals Toolkit total system performance reportsQZRemote seal system performance calculation report HART Revision configuration(10)HR 5(14)Configured for HART Revision 5 HR7(15) Configured for HART Revision 7 Typical model number: 2088 G 2 S 22 A 1 B4 Revision 5 is the default HART output.
7 The Rosemount2088 with selectable HART can be factory or field configured to HART Revision 7. To order HART Revision 7 factory configured, add option code of construction comply with recommendations per NACE MR0175/ISO 15156 for sour oil field production environments. Environmental limits apply to certain materials. Consult latest standard for details. Selected materials also conform to NACE MR0103 for sour refining available with Low Power transmitter output code available with Alloy C-276, materials of construction code of a 1/2 14 NPT housing with a G1/2 1/2 14 NPT female process connection code Assemble-to items are specified separately and require a completed model available with conduit thread code available with 4 20 mA HART output (output code A). configuration buttons (option code D4 or DZ) or local operator interface (option code M4) if local configuration buttons are shipped with 316 SST conduit plug (uninstalled) in place of standard carbon steel conduit transmitter output code S with either materials of construction code 22 or materials of construction code 22 with process connection code the HART output to HART Revision 5.
8 The device can be field configured to HART Revision 7 if the HART output to HART Revision 7. The device can be field configured to HART Revision 5 if needed. Table 1. Rosemount 2088 pressure transmitter Ordering InformationThe starred offerings ( ) represent the most common options and should be selected for best delivery. The non-starred offerings are subject to additional delivery lead 2088 September specificationsFor zero-based spans, reference conditions, silicone oil fill, 316L SST isolating diaphragmReference accuracy of calibrated span. Includes combined effects of linearity, hysteresis, and repeatability of calibrated span (high accuracy option P8)For spans less than 10:1, accuracy = % of spanAmbient temperature effectExpressed as a total effect per 50 F (28 C)Total effect includes zero and span effects ( URL + of span)StabilityRanges 2 4: of URL for 3 yearsRange 1: of URL for 1 yearVibration effectLess than of URL when tested per the requirements of IEC60770-1 field or pipeline with high vibration level (10 60 Hz mm displacement peak amplitude/60 2000 Hz 3g)Power supply effectLess than of calibrated span per volt change in voltage at the transmitter terminalsMounting position effectZero shifts to inH2O (6,22 mbar), which can be zeroed Span.
9 No effectTransient protectionTested in accordance with IEEE , Location Category B6 kV crest ( s 100 kHz)3 kA crest (8 20 microseconds)6 kV crest ( 50 microseconds)General specificationsTested to IEC 801-3 Warrant y(1)All Rosemount 2088 models: One-year limited warranty is standard(2) Extended three-year and five-year limited warranties available if ordered.(3) (Select option WR3 or WR5 in model string respectively.) details can be found in Emerson Terms & Conditions of Sale, document 63445, Rev G (10/6). are warranted for 12 months from the date of initial installation or 18 months from the date of shipment by seller, whichever period expires and five-year warranty period begins on date of shipment by Electromagnetic compatibility (EMC)Meets all industrial environment requirements of EN61326 and NAMUR NE-21(1). Maximum deviation < 1% span during EMC disturbance(2).
10 NE-21 does not apply to Low Power ( transmitter output option code N). surge event device may exceed maximum EMC deviation limit or reset; however, device will self-recover and return to normal operation within specified start-up 2088 September specificationsOutputCode S: 4 20 mACode N: 1 5 Vdc, Low PowerOutputs are directly proportional to the input pressureSelectable HARTD igital communications based on HART Revision 5 (default) or Revision 7 (option code HR7) protocol can be selected. The HART revision can be switched in the field using any HART based configuration tool or the optional local operator interface (LOI).ServiceLiquid, gas, and vapor applicationsPower supplyExternal power supply required. transmitter operates on Vdc with no load ( V for Low Power). Reverse polarity protection is limitationsReverse polarity protection is standard. Maximum loop resistance is determined by the power supply voltage as described by the following equations:Figure 1.